Two independent dose calculation methods have been explored to validate MLC-based IMRT plans from the NOMOS CORVUS system. After the plan is generated on the CORVUS planning system, the beam parameters are imported into an independent workstation. The beam parameters consist of intensity maps at each gantry angle. In addition, CT scans of the patient are imported into the independent workstation to obtain the external contour of the patient. The coordinate system is defined relative to the alignment point chosen in the CORVUS plan. The 2 independent calculation methods are based on a pencil beam kernel convolution and a Clarkson-type differential scatter summation, respectively. The pencil beam data for a 1 × 1-cm beam, as formed by the multileaf collimator, were measured for the 6-MV photon beam from a Siemens PRIMUS linear accelerator using film dosimetry. In the pencil beam method, the dose at a point is calculated using the depth and off-axis distance from a given pencil beam, corrected for beam intensity. The scatter summation method used the conversion of measured depth dose data into scatter maximum ratios. In this method, the differential scatter from each pencil beam is corrected for the beam intensity. Isodose distributions were generated using the independent dose calculations and compared to the CORVUS plans. Although isodose distributions from both methods show good agreement with the CORVUS plan, our implementation of the differential scatter summation approach seems more favorable. The 2 independent dose calculation algorithms are described in this paper.
- CORVUS system
- Dose calculation algorithm
- Photon beam
ASJC Scopus subject areas
- Radiological and Ultrasound Technology
- Radiology Nuclear Medicine and imaging